Features: Fast test time:: test takes approximately 5 seconds:: Accurate1 testing Only a small 1.5?L blood sample:: Capillary fill function. For fast:: easy blood application. Expired test strip warning:: Alarm clock function reminding you to test:: Target range indicator. A warning indicator that will flag when a result is outside the target range. Hygienic test strip ejector. The test strip ejector helps to avoid the handling of used test strips. 300 test memory with time and date so you can analyse results. Discreet:: ergonomic design allows this meter to fit comfortably in the palm of the hand. Larger buttons allow for easy scrolling. Specifications: Memory capacity- 300 results with date & time. 7:: 14 and 30 day test averages. Alarms: 4 selectable alarm times. Underdosing detection: Yes:: Re-dosing is possible within 15 seconds:: Meter set up: Standard and quick set-up:: Buttons: Three: two arrow buttons for easy scrolling:: one ON/OFF button. Test measurement principle: Photometric measurement of the intensity of the colour formed in the reaction. Then conversion to the corresponding glucose concentration. Measuring time: Approximately 5 seconds. Hypo warning: Target range warning (2 thresholds). Calibration coding: Via code chip. Battery: One 3-volt coin cell battery (CR2032):: pre-installed. Battery life: Approximately 1000 readings or 1 year (based on 3 daily readings) Auto power: Auto-on through strip insertion:: Auto-off through eject function. Meter shuts off 60 or 90 seconds after last interaction:: Strip expiry warning: Yes with time and date programmed. Strip ejector: Yes:: for hygienic disposal of used test strips. Acoustic mode: Yes:: can be turned on by user. Flagging of results: Yes:: to indicate special occasions:: e.g.:: AST and quality control tests. Measuring range: 0.6-33.3 mmol/L Storage conditions: -25?C to +70?C without battery:: -10?C to +50?C with battery:: System operating temperature: +10?C to +40?C:: Display: liquid crystal display (LCD) with icons:: Relative humidity operating range: Less than 85% relative humidity:: Operating altitude: Sea level to 4000 m:: Meter dimensions: 102 x 48 x 20 mm:: Meter weight: Approximately 57 g with battery How to Use : Testing Blood Glucose:: Wash your hands thoroughly with warm water and dry them well:: Insert a test strip into the meter to turn on the meter. Next:: verify the code number. After the code number has been displayed:: the test strip symbol with a flashing blood drop symbol appears. At this point:: the acoustic signal sounds:: if it is turned on. The blood drop symbol indicates the meter is ready to perform a blood glucose test. With a lancing device such as Accu-Chek Softclix lance the side of your fingertip (or alternate site if appropriate). Touch the tip of the test strip to a small blood sample. The hour glass symbol is displayed and the meter beeps (only if the beep tone has been turned on). After approximately 5 seconds:: the measured result is displayed and the meter beeps (only if the beep tone has been turned on). The result is automatically stored with time and date in the memory. For more information on how to use your meter:: please see your Accu-Chek Go Owner?s Booklet.
ACCU-CHEK Go Glucose Monitor Only Glucometer(Black) - Specifications
Number of Strips
1 Log book, 1 User manual, 1 Warranty card, 1 Accu-Chek Go Meter (with 1 enclosed battery), 2 Information Sheets, 1 Quick Reference Guide, 1 Carry case